?At best, a thermometer only ever measures its own temperature?. The aim is therefore to equalise the temperature of the sensor element compared to that of the medium to be measured. Only then will the thermometer display the actual medium temperature.
Principles to be considered
The heat always flows from the warmer body to the colder one
Bodies at different temperatures shoot for a balance of the temperature difference
Heat flow from a warmer to a colder body
Striving for a balance of the temperature difference
Heat transport between thermometer and environment
A thermometer transports heat via its thermowell and its own internal components to the surroundings. Here, the assumption is that the ambient temperature is leaner compared to the medium temperature. In the converse situation, the procedure medium is heated by the thermometer.
In general which means that each thermometer generates a heat dissipation. The art is to minimise the resulting error.
digital pressure gauge affecting the magnitude of the heat dissipation
Temperature difference between the medium to be measured and the ambient temperature of the thermometer
Heat capacity of the medium to be measured
Heat capacity of the thermometer (and its components)
Heat penetration coefficient of the material of the thermometer
Thermal conductivity of the thermometer (and its components)
Mass ratios (thermowell, neck tube, medium to be measured)
The physical design of the thermometer is, in the ultimate analysis, the consideration of the sum of all the mentioned influencing factors.
Thermometer components with high heat dissipation
With electrical thermometers: thermowell, outer sheath of the MI cable, wires
With gas-actuated thermometers: thermowell, stem, capillaries
With bimetal thermometers: thermowell, spindle
The heat dissipation thus occurs predominantly via the metal the different parts of the thermometer; however, the air enclosed in the thermometer also transports heat ? though to a much lesser degree. The heat dissipation is ? for the same material ? also stronger, the larger the effective area is through which heat is transported. The higher the mass of a thermometer and its own thermowell and the higher its thermal conductivity, the higher heat energy is that may be extracted from the measuring point.
Conditions to avoid heat dissipation errors
Exemplory case of a measuring point for which a large heat dissipation/measuring error can be expected
The physical design of the thermometer should be matched to certain requirements of the measuring point (so far as is physically possible).
The thermometer must be immersed sufficiently deep in the medium whose temperature it should measure. If this is not the case, under certain circumstances, so much heat will be transported away in to the surroundings that the sensor will not hold sufficient heat energy to adequately detect the medium temperature. It will then be barely possible to keep up the required class accuracy.
The measuring point should be well insulated so far as possible.
Information on our temperature measuring instruments can be found on the WIKA website.
See also our article:
Temperature ? what happens to be it?